日本JaICA Test kit for Potential Anti Oxidant (PAO)
上海金畔生物作为日本 JaICA 老化中国代理商，欢迎新老客户访问日本 JaICA 老化官网或者咨询我们获取更多日本 JaICA 老化产品线价格说明书等信息。
Suitable for detection of total antioxidant capacity in serum and food extracts. For research use only.
Oxidative stress plays on important role in various diseases and aging. The control of oxidative stress is expected to be useful to prevent diseases and aging.Oxidative stress is caused by the imbalance between reactive oxygen species (ROS) and antioxidant defense system. For accurate assessment of oxidative stress, measurement of ROS, oxidative damage and antioxidant activity may be essential. Recently, antioxidants as functional foods which scavenge ROS attract a great deal of attention.
Principle of this assay:
In the PAO assay kit, an easy and convenient method to measure antioxidant capacity is provided. Utilizing the reduction of cupric ion (Cu++ to Cu+), antioxidant capacity of samples can be detected in 5 minutes. Samples are mixed with Cu++ Solution. Cu++ are reduced by antioxidants to form Cu+. Reduced Cu+ react with Chromatic Solution (Bathocuproine) , and can be detected by absorbance at wavelength 480 to 490 nm. Antioxidant capacity can be calculated from the Cu+ formed. PAO can detect not only hydrophilic antioxidants such as Vitamin C, glutathione, but also can detect hydrophobic antioxidants such as Vitamin E. Applicable for assessment of total antioxidants of serum, foods and beverage samples.
||Colormetric assay(detection: 480 – 492 nm)
||21.9 – 4378 碌mol/L (cupric ion reducing power)
||Room temperature (10 – 25掳C)
||Human and animal serum samples, foods and beverage samples.
|Required but not provided:
||A micro plate reader (measuring wavelength 492 nm)
Pipettes and pipette chips
Plastic test tubes
NaOH, HCl solution and pH meter (Not required if standards are prepared with distilled water only).
Content of this kit: 试剂盒组成
|Standard (Uric acid powder):
|Micro titer plate:
||1 plate (96 wells)
||Prepare 6 levels of standards by diluting 2mM uric acid.
||Please prepare plastic test tubes for 6 levels of standards and each sample. Pour 390 碌L of Sample Diluent, and add 10 碌L of standards or diluted samples.
||Pour 200 碌L of mixture to Micro titer plate. Use 200 碌L of Sample Diluent for blank well.
||Read absorbance at 490 nm (as READ1).
||Add 50 碌L of Cu++solution to each well, mix gently, and incubate at room temperature for 3 minutes.
||Add 50 碌L of Stop solution, mix gently, and read absorbance at 490 nm (as READ2).
||Please draw standard curves by plotting the difference of absorbance readings (READ2 – READ1) as vertical axis, and concentration of uric acid standards (mM) as horizontal axis. Calculate the corresponding uric acid concentration of samples. Multiply corresponding uric acid concentration (mM) of samples by 2189, to estimate antioxidant power (碌mol/L).
1mM of uric acid = 2189 碌mol/L (copper reducing power)
||Oxidative imbalance and cathepsin D changes as peripheral blood biomarkers of Alzheimer disease: A pilot study
E Strafacea, P Matarresea, L Gambardella, R Vona, A Sgadari,MC Silveri, W Malorni
FEBS Letters 579, p2759-766 (2005)
||Oxidative stress and its association with coronary artery disease and different atherogenic risk factors
C. VASSALLE, L. PETROZZI , N. BOTTO, M. G. ANDREASSI and G. C. ZUCCHELLI
Journal of Internal Medicine 256, p308-315(2004)
||Antioxidant capacity as a reliable marker of stress in dairy calves transported by road
P Pregel, E Bollo, FT Cannizzo, B Biolatti, E Contato, and PG Biolatti
Veterinary Record 156, p53-54 (2005)
||Vitamin E-coated dialyzers reduce oxidative stress related proteins and markers in hemodialysis ? a molecular biological approach.
LA Calo, A Naso, E Pagnin, PA Davis, M Castoro, R Corradin, P Riegler, C Cascone, W Huber and A Piccoli
Clinical Nephrology, Vol.62(5), p355-361 (2004)
||Oxidative stress-related factors in Bartter’s and Gitelman9s syndrome: relevance for angiotensin IIsignalling.
Calo LA, Pagnin E, Davis PA, Sartori M, Semplicini A.
Nephrol Dial Transplant 18(8) p1518-1525 (2003)
||Effect of epoetin on HO-1 mRNA level and plasma antioxidants in hemodialysis patients.
Calo LA, Stanic L, Davis PA, Pagnin E, Munaretto G, Fusaro M, Landini S, Semplicini A, Piccoli A.
Int. J Clin. Ther 41(5), p187-192 (2003)
||Restored Antioxidant Capacity Parallels the Immunologic and Virologic Improvement in Children with Perinatal Human Immunodeficiency Virus Infection Receiving Highly Active Antiretroviral Therapy.
M Martino, F Chiarelli, M Moriondo, M Torello, C Azzari, and L Galli
Clinical Immunology 100(1),p82-86 (2001)
|Test kit for Potential Anti Oxidant (PAO)
日本JaICA Test kit for Potential Anti Oxidant (PAO) 说明书 技术资料
Nalgene DS3111-0025 离心管盖，聚丙烯，盖尺寸25mm CENT CLOSURE TUBE PP;25MM Labware and Bottles 20 20 562
Amresco 0258-100G CETYL DIMETHYLETHYL AMMONIUM BROMIDE 十六烷基二甲基乙基溴化铵